Screen Type: High-resolution display
Screen Size: 7 inch (17.7cm) diagonally
Sweep Speed: 25mm/sec
Information: Heart Rate, Lead/Pads, Alarm on/off, Sop2, AED Functions and Prompts, Alarm Selection and Limits, Delivered Energy.
Energy Display: Monitor dispaly indicates both selected and delivered energy
Charge Prompt Type: Voice and visual prompts.
Electrode Impedance Measurement Range: 0-250 ohms.
Patient Connection: 3-lead ECG cable, or 5-lead cable, paddles.
Lead Selection: Display on monitor, paddles, I, II, III, AVR, AVL, AVF, V,
ECG Size: 0.25, 0.5, 1, 1.5, 2, 4cm/mV display on monitor
Heart Rate: 20-300 BPM Heart Rate Alarm, On/Off displayed on monitor, user-selectable.
Smart Alarms: Beeper/Voice prompts indicate shock able rhythm.
Type: Rechargeable, Ni-MH Battery, 12V.
Operating Time: For a new, fully charged battery, 60 defibrillator discharges at maximum energy, or 3 hours minimum of continuous ECG monitoring. Additional parameters will effect operating time with different functions.
Measurement Range: 30~100%, ±2% between 80%~90%, others ±5%
Alarm Range: User set high limit and low limit.
Alarm Accuracy: ±10% within setting values.
Alarm time Accuracy: less than 12 sec.
AED Function: Auto analyze and charge x3 with programmable auto energy level selection, screen prompts, and voice prompts.
Shockable Rhythms: Ventricular fibrillation
QRS complex wave duration
Charge Control: Control on device front panel, press key on paddle.
Energy Selection: Selectable at 2, 5, 7, 10, 20, 30, 50, 70, 100, 150, 200, 300, 360 joules.
Synchronized Mode: Synchronizes defibrillator pulse to patient's R-wave. "SYNC" message displayed on monitor.