Dimension: 270mm*140mm*80mm(D x W x H)
Weight: 3 Kg
Power Adaptor rate voltage:
Internal Battery: 10.8V Rechargeable lithium battery
Power Consumption: 100VA
Maximum Flow Rate: 2100(mL/h)
All the above steps were performed at 0.01ml /h.
0.01~9999.99mL,at step 0.01mL
Purge Rate(bolus Rate)
2mL Syringe: 1~100mL/h;
5mL Syringe: 1~150mL/h;
10mL Syringe: 1~400mL/h;
20mL Syringe: 1~600mL/h;
30mL Syringe: 1~1000mL/h;
50mL Syringe: 1~2100mL/h;
All the above step at 1mL/h.
3mL for Rate≥10ml/h
1mL for 1ml/h ≤ Rate < 10ml/h
KVO Rate = Settinge Rate < 1ml/h
Time setting range
00h00min~99h59min, at step 1min
Maximum:130kPa ± 30kPa, Minimum: 26kPa ±
20kPa,level 9 continuously adjustable,See 7.4.7
for setting of occlusion pressure level
Eight Modes: R+V, R+T, V+T, R, Drug library,
Weight, Relay mode, Gradient mode.
Battery operating time at intermediate rate
Battery can run continuously for more than 3 hours after being fully charged.
a)Carry out data transmission through WiFi with
"WiFi debugging" app;
b)The user access control adopts the account
password system and refuses to log in illegally.
Maximum infusion pressure
Time to activate the occlusion alarm; max bolus (Essential Performance)
Minimum rate: Occlusion alarm pressure is
26kPa±20kPa, time is < 1h; Occlusion alarm
pressure is 130kPa±30kPa, time is <3h30min.
Medium rate: Occlusion alarm pressure is 26kPa ±20kPa, time<15min, bolus volume is less than 1mL; Occlusion alarm pressure is 130kPa±30kPa, time<45min, bolus volume is less than 2mL.
(Create a occlusion point in one meter from outlet of syringe)